CCP QuickRef Handbook
MedSystem III Pump
Watchdog Mode: Blank Screen
- Pump self-diagnostics have detected instrument error condition.
- Attempt pump reset
- Power cycle pump using ON/OFF button
- Press START/STOP button to resume each channel that had been infusing
- If Watchdog mode recurs or pump cannot be turned on, discontinue use and tag out for service.